If you have an eddy-covariance system, you understand the importance of performing routine maintenance tasks so you can continue to acquire high-quality, consistently reliable data. Some of these maintenance tasks are straightforward, such as making sure that the optics on your gas analyzer are clean and that the transducers on your sonic anemometer are clear of any debris. Unfortunately, there’s another common maintenance task that poses some safety challenges: ensuring that the air inside your gas analyzer is free of CO2 and H2O by using sufficient scrubbing agents.
In the past, you’ve probably scrubbed the internal air of your gas analyzer using bottles that contain chemicals such as magnesium perchlorate and DECARBITE®. If you have ever had to replace these chemicals, you have likely faced some of these challenging issues:
At Campbell Scientific, we strive to make top-of-the-line instruments that are easy to maintain and safe for both you and the environment. We are pleased to provide you with a safer scrubbing solution. You can now use our patent pending molecular sieve technology to scrub CO2 and H2O from the internal airspace of all our gas analyzers. The new molecular sieve is a drop-in replacement for our old chemical bottles that were used in the EC150, IRGASON, and EC155 gas analyzers.
With the new, safer molecular sieve bottles, you can expect several benefits.
Recently, we replaced all the internal chemicals in the gas analyzers we manufacture with a 13X zeolite molecular sieve. This molecular sieve is a material with small pores that blocks the passage of large particles such as water vapor and CO2. This design ensures that the air in the internal housing of the gas analyzers is free of those gases. With this patented technology, we are the sole provider of NDIR (non-dispersive mid-infrared) CO2 and H2O gas analyzers that utilize scrubbing agents—without the use of restricted caustic chemicals.
Tests of the scrubbing ability of the molecular sieve show comparable results to those of the previous magnesium perchlorate/DECARBITE® scrubber chemical bottles. The typical scrubbing time for CO2 is less than a few hours, whereas water vapor is typically removed within a day.
Under normal use, you should only have to replace the molecular sieve approximately once every two years. (If the zero-and-span readings have drifted excessively, you will need to replace the sieve sooner.) The replacement bottles are part number 32897 for the IRGASON and EC150, and part number 33384 for the EC155.
It is a quick and easy process to replace the molecular sieve within your Campbell Scientific gas analyzer. The molecular sieve material for the gas analyzers comes pre-packaged in a ready-to-use bottle that can be transferred directly into your instrument. You don’t have to worry about pouring, measuring, or spilling the material. In addition, because of the discontinuation of restricted chemicals, you don’t have to worry about shipping regulations and not being able to acquire the needed chemical bottles for the task.
With the previous chemicals that were used to scrub CO2 and H2O, there was a risk of damaging the analyzer housing if liquid water contacted and reacted with the chemical bottles. This could happen if there was a leak that formed in the housing or if the bottle covers were not tightened properly. The result was—at the very least—an expensive repair along with a risk of exposure to the caustic chemicals when handling the instrument. With the addition of the molecular sieve, these risks have been eliminated.
The quality and reliability of your measurements can be influenced by the measures you take to maintain your eddy-covariance system. We hope that the added benefits of the molecular sieve over traditional scrubbing chemicals make the maintenance task of changing scrubbing bottles both safer and easier for you.
Recommended for You: Review the "Molecular Sieve CO2 and H2O Scrubbing Solution" technical paper.
To order the new molecular sieve bottles, please contact Campbell Scientific, Inc., at 435.227.9090. Request part number 32897 for the IRGASON or EC150, and part number 33384 for the EC155. For more information, please contact a member of our support team at 435.227.9100.
higo connector pvc air hose conveyor roller bracket mosaic tiles supplier swimming pool glass wall crystal glass mosaic tile copper mosaic tiles vitreous mosaic tile heat resistant conveyor belt Electric Power Fittings Eye Bolts Eye Bolt HDPE Pipe Floats high quality toggle switch titanfine nitrile gloves Color Coated Steel Coil, Color coated galvanized steel coil